What is it about:
The wristband is connected to a mobile terminal, both Android and IOS, from which the different reminders can be configured and information about the confirmation of its taking by the user can be collected.
Ability Connect App
Emergency call for IOS devices
Aplicación de localización Life 360
Calendula App for medication reminders
Detector de caídas Elea
SMS alerta Samsung
Leotec SOS button
First Call App
Wiser Launcher App
Doro 8031 Smartphone
Fall detector Angel4
Adapted mouse, AdMouse
The lifevit PR-500 bracelet can be purchased in amazon and through the website of the manufacturer:
The price of the bracelet is € 96.99 with VAT and shipping costs included..
The application can be downloaded for free in its Android version at the following link:
The application can be downloaded for free in its IOS version at the following link:
We have tested the bracelet for PR-500 medication reminders and we tell you our impressions below.
This bracelet has two basic functionalities, on the one hand, a light indicator of variable color and on the other hand a button to confirm the taking of medication.
The bracelet has Bluetooth connectivity to send and receive information about the medication shots to an application that can be used in both Android and IOS.
The bracelet has a battery with an approximate autonomy of between 4 and 5 days.
The package of the product includes a set of colored stickers that will be associated with the medicine boxes whose take is to be programmed.
Mode of use
The mode of use is simple, first, the bracelet is connected via Bluetooth protocol to the Lifevit application. To do this, once the Bluetooth of the phone is activated, the search for device button will be pressed. In parallel, the main button of the wristband will be pressed for 5 seconds until it vibrates indicating that the device has been synchronized. The device message configured correctly will appear in the application.
Once connectivity exists, a new medication shot can be added to the application. To do this, the name, color of the label associated with the medication will be included and the time and days in which the medication intake notice must be activated will be recorded. Additionally you can configure a contact email of the person you want to receive the information regarding the shots.
Once this data is recorded on the wristband, it can now work autonomously without being linked during its operation to the mobile phone and its application. In this way you can place the bracelet on the arm of the person who needs a support to remember their medication. The placement of the bracelet is simple.
This implies that the fact that the user confirms the taking of a medication does not trigger the immediate notification of its taking to the desired contact person. The notification will only be made when the wristband is reconnected via Bluetooth to the mobile phone that has the Lifevit application installed.
To identify the shot, a sticker of the desired color is placed in the box of the medication to be programmed, so that once a light of a certain color is turned on in the bracelet, the user should take a dose of the medication associated with the same color. Once the shot is taken, press the main button of the bracelet so that it is reflected in the system. At this point we note as a recommendation to the manufacturer to enable the possibility of programming several doses of medicines simultaneously in some way since up to now the intakes must be programmed with a certain time difference between them.
About the app
The application has a simple and attractive design and is implemented for Android and IOS.
Once the application is accessed, it allows the connection of different devices such as thermometers, scales or tensiometers that the company distributes. In this case it will be necessary to access the pill reminder section in order to configure the wristband. In this sense we believe that it could be more appropriate to make an application dedicated to the medication bracelet.
Once started there is a small guided step by step through pop-up menus that allows the device configuration.
As mentioned in the section on operation, the introduction of new doses of medicines is simple and once registered in the bracelet they work correctly. The important thing in this case is to make sure that there is Bluetooth connectivity and that the shots are well registered on the wristband.
Characteristics of the device
The bracelet is made of a plastic material similar to any current Smartband model.
The charging connector is not standard, and although it gives rise to confusion, since it has no polarity it can be connected in any of its 2 positions.
Although the product is presented as water resistant in the package, it should be noted that the manual indicates that the bracelet is not ready to be submerged and therefore the user should remove it before entering the shower.
The colored stickers are strong enough to stick to the medicine boxes or pillboxes for a long time.
It is possible to check the battery level by pressing the main button of the wristband continuously. Green or red As we have said during the tests, the battery lasted around 4 days with moderate use.
In short, the bracelet offers an innovative and useful method for the management of medication, generally with older users. In our opinion, the entire communication part of the medication to be taken by the user works satisfactorily and is very useful. It would be convenient to improve the aspects that facilitate the communication of the shots to the family member or caregiver, as well as the management of the medicines to be taken remotely.